What is T.E.N.S.?
The initials T.E.N.S. are short form for Transcutaneous Electric Nerve Stimulation which is what a T.E.N.S. machine actually does. A T.E.N.S. machine stimulates nerves electrically by transmitting current through the skin (Cutaneous tissue of Transcutaneous). Now through modern medical technology it may be possible to relieve some types of pain by utilizing the T.E.N.S. method of durgless pain relief. T.E.N.S. is an adjunct to post traumatic acute pain.
Now, through modern medical technology, electrical stimulation may be used at home conveniently and safely. The T.E. N.S. unit is effective, drugless, painless proven, convenient, portable, and inexpensive. T.E.N.S. is an alternate of chronic and many acute pain conditions. In many clinical studies Transcutaneous Electrical Nerve Stimulation (T.E.N.S.) has proven to be one of the safest and most effective alternative to drugs in the treatment of some pain syndromes.
ND3390 Model E8 T.E.N.S. E8 皮外電療止痛機
A very high quality dual channel constant T.E.N.S. with adjustable pulse rate and width features. Three pulse modes. Normal,Modulate and Burst. Very handy size, it measures only 24 x 67 x 93 mm.
The intensity and pulse rate switches are located on the top of the unit so that the user can adjust it by themselves easily. The pulse mode selection and pulse width switches are located inside the battery compartment. The physician can adjust these selections and close the battery compartment. The battery compartment lid is transparent so that the user can see what selections they have made but without changing it.
||Dual, two output channels for stimulation of four electrodes.
||Normal- Modulate- Burst
||adjustable 0-40 mA r.m.s. peak into 500 ohm loading each channel, constant current.
||0-120V. (open circuit)
||adjustable from 25 microsec to 240 microsec. Pulse width remains adjustable at any mode function setting.
||adjustable from 2 Hz to 160 Hz, pulse rate adjustable on any setting of mode function.
||Biphasic square wave.
||Green lamp to show power on for each channel and pulse rate speed for both channel, and red low battery lamp which is to illuminate when battery power reaches about 5.1V.
||Normal - when mode slide switch is in normal setting pulse rate is adjustable from 2 Hz to 160 Hz.
Modulate - when mode slide switch is in modulate setting pulse rate will modulate ramped down to about 70% from pulse rate setting. Modulate time period is fixed at 6 seconds
Burst - Burst mode automatically bursts on and off two times per second (two bursts per second) at adjustable pulse rate.
||9 volt battery (alkaline, or nickel-cadmium, rechargeable)
||silicone rubber electrode, 4.2 x 4.2 cm
||95 gm (body only, battery and accessories not included).
||24 x 67 x 93 mm
||Steel spring belt clip.
||2 pcs. pin lead wire
4 pcs. silicone rubber electrode
2 pcs. hydro pad
1 pc. instruction manual
1 pc. carrying case conventional