What is T.E.N.S.?
The initials T.E.N.S. are short form for Transcutaneous Electric Nerve Stimulation which is what a T.E.N.S. machine actually does. A T.E.N.S. machine stimulates nerves electrically by transmitting current through the skin (Cutaneous tissue of Transcutaneous). Now through modern medical technology it may be possible to relieve some types of pain by utilizing the T.E.N.S. method of durgless pain relief. T.E.N.S. is an adjunct to post traumatic acute pain.
Now, through modern medical technology, electrical stimulation may be used at home conveniently and safely. The T.E. N.S. unit is effective, drugless, painless proven, convenient, portable, and inexpensive. T.E.N.S. is an alternate of chronic and many acute pain conditions. In many clinical studies Transcutaneous Electrical Nerve Stimulation (T.E.N.S.) has proven to be one of the safest and most effective alternative to drugs in the treatment of some pain syndromes.
NP3633 Model Pro T.E.N.S. PRO 皮外電療止痛機
A dual channel pocket size T.E.N.S. unit, features with pulse width and rate switch, mode selection for normal, burst, and modulate functions. Equipped with a built in adjustable timer so that the unit will shut off at set time. Operated by 9V dry battery.
||Dual, isolated between channels
||Asymetrical bipasic square wave, zero net DC
||0 to 80V each channel, adjustable (500 ohm load) Pulse Frequency :2 to 150 Hz adjustable
||50 to 250 micoseeconds, adjustable
||Pulse amplitude varies in a cyclic pattern over interval of nominally 5 seconds.
Pulse amplitude decreases linearly and softly over a period of 1 second period to its original value. The original pulse amplitude is maintained for 2.5 seconds. The cycle is then repeated.
||7pulses per burst
2 bursts per second
100 Hz internal frequency of burst
||100 volts, open circuit
||16 micro coulombs per pulse
||30 Min, 60 Min or constant time
||9 volt alkaline battery or similar rechargeable cell
||100 hours average usage (alkaline)
||24 x 64 x 95 mm
||130 grams (including battery)
||2 pcs. pin lead wire
4 pcs. silicone rubber electrode
2 pcs. hydro pad
1 pc. instruction manual
1 pc. carrying case